This video will show you how to do a Baseline. This video is for Spooky2 software version 20191122 and above.
Connect the power adapter. Then connect the USB cable.
Connect the Tens cable to Out1 port of G1 or G2. Here I choose G1. Twist to lock the cable. The other side connect to tens pads.
Stick tens pads to where you want to scan. For example, I want to scan my kidney, so I place Tens Pads across both kidneys.
Turn on your generator.
Launch Spooky2 software.
Go to the Presets tab, select /Biofeedback, GeneratorX, and then select a preset you need. We suggest that you run the GX Biofeedback Scan (C) – JW preset if you don’t have special needs.
Then go to the Control tab, tick “Overwrite Generator” and click the generator port of G1. You could see which generator belongs to which port here. It takes about 26 minutes.
Tick “Baseline Before BFB” here. This feature will let the software automatically perform a biofeedback scan after the Baseline has been created.
Click “Scan” to start the Baseline. Now the countdown is going and software will scan frequencies one by one and find out the most suitable frequencies for me.
1.The biofeedback scan will automatically run after the Baseline is created, so you don’t need to do anything after click “Scan”. Just wait for the window of scan results pop up.
2.If you run Sample Digitizer scan, you need to put the samples in Spooky2 Sample Digitizer at the very beginning.
Thanks for watching this video. Do you have other questions about this new feature? Welcome to tell us!